Measurements of water column depth at 15 minute intervals in Plum Island Sound at the Ipswich Bay Yacht Club, for year 2017. OTT radar level sensor (RLS) installed September 20, 2011 out of the water under the concrete pad on the Ipswich Bay Yacht Club pier, Ipswich, MA. RLS was mounted so that continuous year round measurements can be conducted without the concern of ice flows damaging the sensor.
RLS water level install,IBYC, 9/20/11
Hap Garritt, Sam Kelsey(MBL), Nichole Halsey, RonanO’Maitu (Hach)
Used Growler boat at high tide to attach 2 x 6 pressure treated wood underneath pier platform between pilings. Wood will be used to attach the RLS. Wood attached with concrete adhesive and concrete screws.
RLS attached to wood with screws and leveled with shingle shims.
RLS cable attached to 2x4 wood which was secured along concrete. Cable attached to eye screws using cable ties.
Pier deck to sediment bottom, deck near logger down to bottom by floating dock directly downstream of logger/deck railing = 22.45 ft or 6.84 meters
RLS height above pier deck = 33 inches or 0.838 meters
IBYC NAVD88 benchmark/pier deck by datalogger = 2.78 meters, as determined by Trimble high precision GPS.
Benchmark height checks
Year NAVD88 Comments
2016 2.790 new pier joists and decking at IBYC , late spring 2015
RLS to sediment bottom = 6.84 + .838 = 7.68 meters
RLS relative to NAVD88 = 3.618 meters
The OTT RLS sensor technology is based on impulse radar. Two antennas are enclosed in a compact housing and transmit pulses toward the water surface, through a Teflon face plate. The transmitting antenna transmits short radar pulses in the 24 GHz ISM band. The receiving antenna receives the pulses reflected from the water surface to determine the distance between the face plate of the radar sensor and the water surface. The time delay from transmission to receipt is proportional to the distance between sensor and water surface. Approximately 16 individual measurements are conducted per second and averaged after 20 seconds to minimize wave influences.
There may be sporadic instances of no data (documented in Comments field of the data) due primarily to low battery voltage over several days causing the datalogger to malfunction.
Collection on going
Version 01: January 25, 2018, data and metadata updates to comply with importation to Drupal and LTER PASTA. Used MarcrosExportEML_HTML (working)pie_excel2007_Jun2017.xlsm 6/15/17 10:23 AM for QA/QC to EML 2.1.0
Version 02: February 8, 2018, corrected time variable in MON-SO-IBYCRLS2017.xls, Excel file (had a 2016 date). Used MarcrosExportEML_HTML (working)pie_excel2007_Jan2018.xlsm 1/30/2018 2:24 PM for QA/QC to EML 2.1.0